multi-reflector Surgery Shadow less lamp
Under shadowless lamps, the light energy increased with low-magnification loupes and decreased with high-magnification loupes. Filtering eyeglasses reduced the energy, especially in conditions where the low-magnification loupe was used. The best color-rendering index values were obtained with computer eyeglasses under conventional light-emitting diode shadowless lamps and with no glass and with lightly yellow-tinted lenses under less-blue light-emitting diode.
Color Temperature(k): 4500±500
Color rendering index: 100≥Ra≥85
Diameter of Spot(cm): 10-28
Depth of light beam(mm): ≥700
Brightness level adjustment: CPU light-dimmer
The value of the total irradiance: <6Mw/m²lx
(service range for temperature rise≤10℃)
Total irradiance: ≤6Mw/m²
(Temperature rise on the head of patient under surgical operation≤2℃)
Diameter of lamp holder(mm): 500
Shift of major and stand by lamps: Automatic
Bulb power W(24V): 150
Power supply (AC): 220±20% 50Hz
Input power(VA): 200+10%
1. Use –Examination of wounds and provision of additional light for inserting needles.
2. Unique optical system, homogenous light spot and high-definition.
3. 360 degrees rotation for all joints of rotator, balance arm, operating lamp.